CMS60D Pulse Oximeter adopts Photoelectric Oxyhemoglobin Inspection Technology in accordance with Capacity Pulse Scanning & Recording Technology, which can be used to measure SpO2 and pulse by finger. It is applicable for use in hospital, oxygen bar, community healthcare and exercise healthcare(It can be used before or after exercise, but not recommended to use during exercising).
1.Small in volume, light in weight and convenient to carry.
2.Easy and simple to operate, low in power consumption.
3.With function menu.
4.Display of SpO2 value.
5.Display of PR value and bar graph.
6.Display of pulse waveform.
7.Adjustable screen brightness, PR sound indication.
10.Audible prompt for over-limit and low battery.
11.Battery status indication.
12.Low battery indication.
13.Storage function for SpO2 and PR values, and data stored can be uploaded to PC.
14.Real-time data can be uploaded to PC.
15.External SpO2 probe can be connected.
Display mode: color display
SpO2 measurement range: 0 % ~ 100 %(resolution: 1 %)
Accuracy: 70%~100%: ±2%
PR measurement range: 30 bpm~250 bpm(resolution: 1 bpm)
Accuracy: ±2 bpm or ±2 %, whichever is greater.
Measurement performance in weak filling condition: SpO2 and pulse rate can be shown correctly when pulse-filling ratio is 0.4 %. SpO2 error is ±4 %, pulse rate error is ±2 bpm or ±2 %, whichever is greater.
Resistance to surrounding light: The deviation between the value measured in the condition of man-made light or indoor natural light and that of darkroom is less than ±1 %.
Power consumption: ≤ 100 mA
Voltage: DC 3.0 V
Power supply: two AA batteries
Continuous working time: 44 hours in theory
Safety class: internally powered equipment, type BF applied part
Dimension: 110 mm(L) × 60 mm(W) × 24 mm(H)
Weight: about 120 g(including battery)
Payment & Security
Your payment information is processed securely. We do not store credit card details nor have access to your credit card information.